HAMEG INSTRUMENTS HM | Module: function generator; 7-Segment,5- digit LED; Band: ≤10MHz – This product is available in Transfer Multisort. F requency Range 50 mHz 10 MHz,. Output Voltage up to 10 Vpp (into 50 Ω). ☑ Waveforms: Sine Wave, Triangle, Square Wave, Pulse, DC. ☑ D istortion Factor. Funktionsgenerator HM Equipment HAMEG GmbH, Frankfurt, build ?, 4 pictures, 4 schematics, Germany, tubes, semiconductors, Service- or Lab.
|Genre:||Health and Food|
|Published (Last):||25 March 2010|
|PDF File Size:||8.40 Mb|
|ePub File Size:||4.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
Instead, contact your nearest service center! Suritett levegovel hameb a port valamennyire. If the trigger output is loaded with 50 ohms, the value falls far below the TTL level. In addition, some files are archived, so you need WinZip or WinRar to open that files. If exact square-shaped signals are required, care should be taken that only 50 ohms coaxial cables e. Additional quality fea- tures include the relatively low distortion factor of the generated signals and constant amplitude flatness throughout the entire frequency 8300 of the instrument.
Adjust 0 for minimum overshoot or rounding, respec- tively, hameh the rising edge. By clicking the accept button below, you agree to the following terms.
After unpacking the module, check for any mechanical damage or loose parts inside. HAMEG is not liable for hamge dam- ages. Vasaroltam egy hameg es szkopot. The logical front-panel layout of the module hamef rapid familiarisation with the various functions. In the open-circuit condition, the available signal amplitude is about twice as high. The Manufacturers and RS disclaim all warranties including implied warranties of merchantability or fitness for a particular purpose and are not liable for any damages arising from your use of or your inability hameh use the Information downloaded from this website.
First-time operation of the module Provided that all hints given in the operating instructions of the HM Mainframe were followed – especially for the selection of the correct mains voltage – start of operation consists practically of inserting the module into the right or left opening of the mainframe. An offset voltage adjusted at the 50 ohms output has no influence upon the trigger signal.
If the output of the HM unit comes into contact with components of the circuit under test, which are carrying DC voltage i. A small circle o is now revealed on the red power button in the front centre of the mainframe. Connect the oscilloscope to output. Select a screen height of hzmeg divisions.
HM Hameg | HM function generator module,5MHz | | RS Components
H 68, D mm Weight: Corres- ponding to Safety Class 1 regulations three-conductor AC power cable. Output attenuator function Setting: The instrument should be returned in its original packaging for maximum protection.
When later replacing the case, care should be taken to ensure that it properly fits under the edges of the front and rear panels. The specified adjustment sequence should be strictly observed. The output impedance is 50 Q, and the max. In case of cleaning the module the red display pane should not come in contact with alcohol or methylated spirit.
The selected frequency appears on the 4-digit display 0. The obtained results can be examined by use of the formulas specified in the “FM input” section of the operating instructions.
To obtain the normal operating temperature, the mainframe with inserted module should be turned on at least 30 minutes before hsmeg the test. The TP 1 test point is located next to the amplitude control on the upper PC board. Please supply name and department, address and telephone number to ensure rapid service.
The mains plug of the HM should be inserted before connections are made to measuring circuits. Before opening the set, you should carefully read the Safety hamey Warranty information on page M2 of these operating instructions.
Signal waveform at the trigger output Setting: It should be noted that only the displayed digits are valid; the decimal point is not taken into consideration e. A negative DC voltage reduces the frequency.
Maintenance The most important characteristics of the module should be periodically checked.
You have chosen to save the following item to a parts list:. Maximum output amplitude Setting: The product detailed below complies with the specifications published by RS Components.
Remove both screws on rear panel and, holding case firmly in place, pull chassis forward out of case. Before exchanging the module, the mainframe must be switched off.